# Difference between revisions of "Switching Modes"

To switch between Integrating Mode and Counting Mode for the PREX DAQ.
PREX Main<< HOW TOs

This page details the checklists for switching between counting and integrating modes. For reference, here are the instructions from PREX I. It is critical to do these steps in the correct order, especially for switching from counting to integrating. The detectors which are used in counting mode can be damaged if they are on when the rate is as high as it will be in integrating mode.

1. Quartz HV are first 2 channels of S1 (2nd card)
2. L-HRS quartz rpi8 .... L1.0 = -905.0; L1.1 = -0; L1.2 = -1100.0; L1.3 = -1100.0 .... integrating mode
3. R-HRS quartz rpi4 .... L1.0 = -910; L1.1 = -0; L1.2 = -950.0; L1.3 = -1000.0 .... integrating mode
4. For counting mode voltages are
1. RHRS (rip4 - S0 slot)
• S0A: L0.0 = -1600 V
• S0B: L0.1 = -1600 V
• S3_1: L0.2 = -1300 V
• S3_2: L0.3 = -1300 V
• S3_3: L0.4 = -1300 V
2. LHRS (rpi8 -S0 slot)
• S0A: L0.0 = -1400 V
• S0B: L0.1 = -1600 V
• S3_1: L0.2 = -1300 V
• S3_2: L0.3 = -1300 V
• S3_3: L0.4 = -1300 V
1. LHRS VDC:- rpi8 - Slot S7 - channels - L6.0 = -3500 & L6.3 = -3500.0 Volt
2. RHRS VDC:- rpi4 - Slot S14 - channels - L14.5 = -3500 & L14.7 = -3500.0 Volt
1. For the S0 and S3 detectors, we should simply need to turn them on (counting mode) or off (integrating).

## Counting to Integrating

1. Turning OFF the HV
• VDC HV
• Scintillator HV
• Adjust Quartz Detector HV for Integrating mode
2. Switching to High-Current Integrating Mode
• Take access.
3. Set Up Beam
4. Take data !

## Integrating to Counting

1. Preliminaries. to do long before switching
• Make sure spectrometer DAQ works (might need reboots). Use the CODA guide if necessary.
• Check with spot++ that rastered size is approx 4 x 3 mm (whatever is our present norm.)
• Check the beam position ("centered" to within 0.3 mm, using the production criterea, see white board).
2. Switching to Low-Current Counting Mode
• Take access.
• Expert to do the switch of the detector signals to counting mode
3. Set Up Beam
• Do the beam recovery procedure.
• Establish 0.3 microA on Ca target (or appropriate current on another target).
4. Setting HV
• Do NOT turn on scint or VDC HV yet ! Must establish rates first.
• Turn on the HV for upper scint and S0 (if S0 is in place).
• Adjust Quartz Detector HV for Counting mode.
• VDC HV on (3.5 kV) if the T1/T2/T6 rates are below 500 kHz. If above 500 kHz do not turn on HV.
5. Take CODA run and analyze it as usual.